Surge tester built to the requirements of IEC 60601-1:2005 (Issue 3) Figure 10 (IEC 60601-1 Figure 50 in older versions). Conducts IEC 60601-1:2005 Figure 11; IEC 60601-2-34 Issue 2(Optional); IEC60601- 2-34 Issue 3(Included); IEC 60601-2-49; EC-13; EC-53 and the new requirement of IEC 60601-2- 25:2011 and IEC 60601-2-27:2011 including Correction 1. Tests of protection against the effects of defibrillation (Common mode, Differential mode), Energy reduction test. All these tests use the same 5000V source with difference in output wave shaping. The various wave shaping output circuits required for each test are included in the D5-PF.
The more remarkable changes of the D5-PF version are: output voltage selected digitally, direct joules readout, increased accuracy, faster time between pulses "20 seconds", increased signal generator voltage, pass/fail reference is integrated, dry cap is used and USB Testminder is standard.
|Common Standards||IEC 60601-1:2005 figure 10; IEC60601-1:2005 figure 11; IEC 60601-2-34,2-49; EC-13 and EC-53|
|Voltage Source||0-5100 V ± 1%|
|Main Capacitance||32µF ± 5%|
|Inductances||500µH and 25µH|
|Duty Cycle||20s between pulses, ± 1% 100Ω resistor bank, up to 70 pulses
12s between pulses, +1%<-5% 100Ω resistor bank, continuous
|Energy Measurement||Works only when 25mH inductor selected
Display Resolution: 1 Joule
Repeatability: ± 1 Joule between pulses
Rent EMC/EMI Test Systems
Get fast deliveries and ISO 17025 calibrations on conducted and radiated rental equipment
US ESD Distributor - Ametek CTS
Buy Teseq & EM Test ESD simulators and test accessories from sole US distributor